Quality flow meter measuring devices provider: Radar level meter adopts non-contact measurement, and has the advantages of high measurement accuracy, not affected by process conditions, easy to install and easy to maintain, so it is widely used in offshore oil platforms in recent years. Today, we would like to take you through several actual cases in offshore oil platform to understand the reasons and solutions of the radar level meter’s measurement value jumping. The first case was an oil field where the radar level meter in the left crude tank jumped to full scale several times during production. A sealed metal cover was used to cover the main body of the sensor, and during the test, it was found by connecting the radar level transmitter that the closed metal cover would result in more false signals throughout the measurement range, and the overall noise line would rise. Read extra information at flow instrumentation.
Data Analytics – IIoT devices generate large amounts of data. Radar sensors equipped with IIoT capabilities often include built-in analytics tools that seamlessly integrate with industrial data analytics platforms. This empowers organizations to extract insights from collected data enabling informed decision making. Enhanced Connectivity – To fully participate in the IIoT ecosystem, radar sensors are equipped with versatile connectivity options such as Ethernet, Wi-Fi or cellular networks. These connectivity options ensure integration with existing control and data acquisition (SCADA) systems or cloud based platforms.
Power supply and output signal, the power supply has 220 V A C, 24 VDC, and the two-wire or four-wire system can be selected according to the needs. Output signal 4 ~ 20 mA DC or digital signal, select the appropriate model according to the needs. The above are all factors that need to be considered when selecting a radar level transmitter. The essence of the details is a serious attitude and a scientific spirit. I hope that every user can choose a suitable radar level meter.
So what are the installation technical requirements for radar water level meters? The installation of the radar water level meter must be vertical to the object to be measured; there should be no obstructions between the measured object and the radar water level meter, otherwise it will affect the reflection of radar waves, that is, affect the measurement accuracy; the center of the radar water level meter is far from the shore of the water body. The distance must be greater than the radius of the transmitting beam, otherwise the measurement accuracy will also be affected; the installation cantilever bracket of the radar water level meter must be firm, and cannot be shaken up and down; in order to protect the radar water level meter, the water level meter can be installed on the cantilever bracket. Iron box, put the radar water level meter probe.
For more accurate measurement in deaerators, magnetrol guided wave radar (GWR) is a preferred option. Since its performance and accuracy are not contingent on the specific gravity and/or inference, it can provide reliable measurements in all situations, including the difficult and turbulent process conditions of deaerators and feedwater heaters. In addition, GWR does not require external inputs or calibration to achieve specified performance. This effectively eliminates the introduction of errors during the calibration process or from external sources, i.e., pressure and temperature. With this high level of accuracy, operators can trust that their deaerators will be well controlled.
Secondly, in cement production, material level measurement encountered another difficulty is strong dust interference, especially pneumatic conveying powder silo, dust flying when feeding, low visibility, laser level meter and high energy ultrasonic level meter can not be measured, although radar level meter at this time can receive part of the surface echo. But the echo signal was also weakened. In addition, due to the uneven surface of the radar echo is refracted, will also lead to the existence of radar echo; In addition, there are some conditions from the bottom to the bin inflation, so that the material surface loose, material level measurement is more difficult.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Discover extra details on https://www.kaidi86.com/. Suitable for chemical industry, petroleum industry, metallurgical industry, water conservancy and electronic industry, etc.
The radar level gauge works by electromagnetic waves. Its working principle is to measure the specific liquid level by transmitting electromagnetic waves to the measured target. After the electromagnetic waves are emitted, they are reflected by the medium. For the radar level gauge, its key function is to ensure that it can transmit magnetrol guided wave radar signals smoothly. In our industrial production site, interference often occurs, so which interference sources will affect the measurement of the radar level gauge? let’s see.
For the continuous measurement of corrosive, high viscosity, flammable and toxic liquids, liquid-liquid interface and solid-liquid interface that are difficult to measure by ordinary liquid level meter , ultrasonic level meter should be selected, but should not be used in occasions with large liquid level fluctuations . Ultrasonic level meter are suitable for measuring media that can fully reflect and transmit sound waves, but should not be used in vacuum. The ultrasonic flow meter should not be used in process equipment where internal obstacles affect the propagation of sound waves.
Working principle: Tube in the liquid level meter has a magnetic float, float design, according to the proportion of medium float magnet steel and container liquid level at the same level, wholesale liquid level gauge in a semiconductor magnetel liquid level gauge sensor LED display scale, and orderly arranged from top to bottom, and LED module corresponding to the magnetic field magnetic sensor, the magnetic semiconductor, triggering the corresponding digital circuit, liquid level above the red LED display, The LED below the liquid level gauge is displayed in green, the red in gas phase, and the green in liquid phase. The red-green junction is the actual liquid level in the container.